|Place of Origin:||China|
|Minimum Order Quantity:||1 Set|
|Packaging Details:||Fumigation-free plywood wooden case for export|
|Delivery Time:||1~30 work days after order confirmed|
|Payment Terms:||T/T, L/C, Western Union, MoneyGram|
|Supply Ability:||20 sets per month|
|Type:||Motor Vibration Type||Power Source:||Single Vibrating Motor|
|Feeding Capacity:||200g||Purpose:||For Both Dry And Wet Screening|
|Voltage:||220V, 50HZ/60HZ, Single Phase||Sieve Dia.:||75mm, 100mm, 200mm|
|Vibration Frequency:||1420 Rounds Per Minute||Sieve Number:||1~8 Pcs Of Sieves|
laboratory sieve shaker machine,
lab sieve shaker
Formal Analysis Test Sieve Shaker , Dry And Wet Table Top Sieve Shaker
Function and Principle:
The AT-300 Formal Analysis Vibration Test Sieve Shaker for Both Dry and Wet Separating is a new type of AAREAL power generator is used to produce multiple high-frequency vibration, which can sieve and filter samples efficiently. A standard analytical sieve that combines powder classification and liquid filtration. It is suitable for the accurate analysis of particle size and powder structure in laboratory and quality control room. Low noise, high efficiency and high precision.
This test screen shaker uses single vibration motor as the vibration source. The motor vibration drives the base of the sieve shaker to vibrate, and then passes through the base to the standard test sieve placed on the base. The material particles smaller than the mesh size of the test sieve are screened into the lower test sieve. Screening layer by layer like this, leaving only the same size of material in each test sieve, the separation of different particle sizes can be achieved and the particle size composition of this material can be determined. Thereby, the material in the test sieve can be effectively sieved, classified and filtered.
1. High quality stainless steel sieve frame, much durable.
2. Sieve frame coupled with each other, no material leakage.
3. Timer design, set testing time as you need.
4. Silica gel cushion, reduce noise, much stable.
5. Small size, do not occupy space, easy to move.
6. Timing automatic shutdown.
7. The sieving machine can be placed up to eight layers of test sieves (including screen bottom), which can accurately separate 2~7 particles at one time.
8. Multi-layer design, convenient dismounting device.
Technical Parameters of Economic Single Vibration Motor Test Sieve Shaker for Both Dry and Wet Sieving:
|1||Screen Dia.||Ø200, Ø100, Ø75||mm|
|8||Power Supply||220; 50||V; HZ|
1. The standard test sieve used in laboratory, and quality control inspection departments such as room of particles, powder material particle size distribution measurement, the product impurities, liquid solids content determination of analysis.
2. Place the standard test sieves in the device and lock the nuts on the top and sides. The method is as follows: first, loosen the four locking nuts on the top and both sides of the standard vibrating screen, so that the two vertical columns on both sides can fall freely, falling on the standard test sieve that has been placed, then lock the locking nuts on both sides, and finally lock the two nuts on the top of the screen machine.
3. If you need to adjust the vibration amplitude, you can adjust the angle between the upper and lower excitation blocks of the motor in the device to a suitable angle to obtain different exciting forces. Generally do not adjust.
4. After doing the above steps, open the green button on the front panel of the standard vibrating screen machine, and after a period of time screening, the screen machine will stop automatically when the time is over, or you can press the red button to end the screening. At the end of the screening, you can screen out the materials of the multiple size segments you need.
5. Adjust the time according to the screening needs. The first two digits of the time relay in the middle of the front panel of the standard vibrating screen are minutes, and the last two digits are seconds. The timing can be adjusted as needed within 0-99 minutes and 59 seconds.
Note: nuts on top and sides must be locked.
Installation and Adjustment Instructions:
1. Pre-operation check
I) If the machine body is properly placed;
II) If the standard test screen is installed in coordination with the standard vibrating sieve shaker;
III) If the top and side nuts of the standard vibrating screen are locked;
IV) If the machine has any contact with other objects.
2. Matters needing attention in operation
I) If there is abnormal noise;
II) If there are any other abnormalities.
3. Daily maintenance
(I) Before starting: (A) Pay attention to the presence of abnormal murmurs;
(B) Check whether the vibration is abnormal or not.
(C) Check whether the current is stable;
(II) After use: Clean up after each use.
The motor vibration type test sieve shaker for both dry and wet separating is widely used in food, medicine, chemical, abrasive, paint, mining, metallurgy, geology, ceramics, defense and other industries of scientific research units, laboratories, inspection room, production control for solid, powder, seriflux and other materials of testing, screening, grading, particle size analysis, impurity content analysis and so on.
Do not debug with power. Adjust it after shutdown.
Contact Person: Ms. Magie